锥管等主部件均由AS、ABS塑料制成，有较好的耐腐性能，产品还具有结构合理、体积小、重量轻、锥管不易破碎等特点，塑料管转子流量计可广泛应用于化工、环保、食品等工业部门。 Plastic pipe rotor flowmeter cone and other main components are made of AS, ABS plastic, have good corrosion resistance, the product also has reasonable structure, small size, light weight, cone tube is not easy to break, etc. Design can be widely used in chemical, environmental protection, food and other industrial sectors.
Principles and Structure
The main measuring element of the plastic tube rotor flowmeter is a conical plastic tube with a small end facing downward and a large end facing upward, and a float that can move up and down inside. When the fluid flows through the conical plastic tube from bottom to top, a pressure difference is generated between the upper and lower parts of the float, and the float rises under the action of this differential pressure. When the force that causes the float to rise is equal to the combined forces of gravity, buoyancy, and viscous force, the float is in an equilibrium position. Therefore, there is a certain proportional relationship between the flow rate of the fluid flowing through the flowmeter and the rising height of the float, that is, the flow area of the flowmeter. The position height of the float can be used as a flow measurement.
安装前，应把仪表内的泡沫支撑物用工具尖嘴钳轻轻把它取出，然后重新把它安装好，在安装时，要注意橡胶垫圈是否装在接管的槽内，以防漏水。  Before installing the plastic tube float flowmeter , the foam support in the meter should be gently removed with tool-nosed pliers, and then reinstalled. When installing, pay attention to whether the rubber gasket is installed on the pipe. Inside the tank to prevent water leakage.  The interface of the flowmeter can be matched with ABS plastic pipe, or PVC plastic pipe, and connected with special glue, or it can be connected by plastic method, and then connected to the metal flange. 3) The flowmeter must be installed vertically, without obvious tilt, and its installation height should be convenient for reading the float. When reading, the line of sight and the float should be horizontal. [General standard: The optimal flow rate required is 50% -70% of the maximum flow rate of the flow meter. Try not to use the highest number.  The upstream and downstream inlets and outlets of the meter should ensure a straight pipe section with 5 times the meter diameter.  The pressure of the bedding fluid must be stable. If the fluid is too unstable, it will cause the float to fluctuate and cannot be measured correctly. A buffer or a setter should be set upstream of the flowmeter.  The plastic tube rotor flowmeter should be installed vertically, and the angle between the centerline of the flowmeter and the plumb line should not exceed 5 degrees.
 When using the meter, first slowly open the upstream valve to full open, and then use the regulating valve downstream of the flow meter to adjust the flow. When the meter is out of use, close the upstream valve slowly, and then close the flow regulating valve downstream of the meter.  The indication value of the meter is seen in the middle of the score line, and the indication value is read from the scale of the float according to the height of the float. 〔3〕 When the cone and float are contaminated, they should be cleaned in time.  Regularly check the flowmeter for errors. Once it exceeds the specified error, the float and cone should be replaced or recalibrated to obtain better accuracy.  When the measured fluid is inconsistent with the calibration medium of the flowmeter, the displayed value must be corrected. It is expressed by the following formula: QS = QN QS = Actual flow rate Qn = Flowmeter reading indication value ρn = Medium density at the time of flowmeter calibration ρs = Density of the measured medium ρf = Float material density
塑料管浮子流量计金属管转子流量计技术参数测量范围:水（20℃）1-200000 l／h 空气（20℃，0．1013MPa）0．03－4000m3／h 参见流量表，特殊流量可订制量程比：标准型10：1 特殊型20：1精度： 标准型1．5级特殊型1．0级压力等级：标准型： DN15－DN50 4.0MPa DN80-DN200 1.6MPa特殊型： DN15－DN50 25MPa ?DN80-DN200 16MPa夹套的压力等级为1．6MPa特殊型在选型和订货前应与工厂协商压力损失：7kPa-70kPa介质温度：标准型：－80℃-＋200℃：PTFE：0℃－85℃高温型：最高可达400℃介质粘度：DN15: <5mPa.s(F15.1-F15.3)<30mPa.s(F15.4-F15.8)DN25: <250mPFa.sDN50-DN150: <300mPa.s环境温度：液晶型－30℃－+85℃指针型－40℃－+120℃连接形式：标准型：DIN2501标准法兰特殊型：由用户指定的任意标准法兰或螺纹电缆接口：M20＊1．5供电电源：标准型24VDC二线制4－20mA（10．8VDC－36VDC）交流型：85－2 Plastic tube float flow meter Plastic tube float flow meter Metal tube rotor flow meter Technical parameters Measurement range: Water (20 ° C) 1-200000 l / h Air (20 ° C, 0.113MPa) 0.03-4000m3 / h See flow meter , Special flow can be customized range ratio: standard type 10: 1 special type 20: 1 accuracy: standard type 1.5 special type 1.0 pressure level: standard type: DN15-DN50 4.0MPa DN80-DN200 1.6MPa special Type: DN15-DN50 25MPa? DN80-DN200 16MPa jacket pressure level is 1.6MPa Special type should be negotiated with the factory before selection and order Pressure loss: 7kPa-70kPa Medium temperature: Standard type: -80 ℃-+ 200 ℃ : PTFE: 0 ℃ -85 ℃ High temperature type: up to 400 ℃ Medium viscosity: DN15: <5mPa.s (F15.1-F15.3) <30mPa.s (F15.4-F15.8) DN25: < 250mPFa.sDN50-DN150: <300mPa.s Ambient temperature: LCD type -30 ° C-+ 85 ° C pointer type-40 ° C-+ 120 ° C Connection type: standard type: DIN2501 standard flange special type: any standard specified by the user Flange or threaded cable interface: M20 * 1.5 Power supply: Standard 24VDC two-wire system 4-20mA (10.8VDC-36VDC) AC type: 85-2 65VAC? 50HZ battery type: 3. 6V @ 4AH Ni-MH battery alarm output: upper or lower limit instantaneous flow alarm standard type: open collector output (maximum 100mA @ 30VDC internal impedance 100 ohms) special type: relay output (contact capacity maximum 5A @ 250VAC ) pulse output: cumulative Pulse output with a minimum interval of 50 milliseconds. Liquid crystal display: Instantaneous flow display. Value range: 0-50,000. Cumulative flow display. Value range: 0-99999999. Protection rating: IP65. Explosion-proof mark: Intrinsically safe type iaⅡCT6. Code measuring tube structure ST-LZ Down-in, top-out, down-in, up-out, down-in, up-out, right-in, left-out, left-in, right-out DN25 25 DN50 50 DN80 80 DN100 100 DN150 150 DN200 200 Code additional structure without T jacket type Z damping type G high temperature type Y high pressure type code indicator form code combination M1 local indicator, mechanical indication instantaneous flow M2 power supply type, mechanical Indicate instantaneous flow rate, LCD display instantaneous / accumulated flow rate M3 power supply type, no mechanical indication, LCD display instantaneous / accumulated flow rate code for The method is not limited to M1 indicator A 220VAC, 4-20mA output B battery power supply, no output C 24VDC, two-wire power supply, 4-20mA output D 24VDC, three- or four-wire power supply, 4-20mA output code explosion-proof mark l AniaCT5 square housing d flameproof Diibt4 round housing code alarm or pulse output without alarm or pulse output K1 upper limit alarm or one pulse output K2 lower limit alarm or one pulse output K3 upper and lower limit alarm or dual pulse output