安装必须垂直、以保证浮球组件在主体管内能上下运动自如； 1. The level gauge must be installed vertically to ensure that the float assembly can move up and down freely in the main tube;
之间装截止阀、以便清洗和检修液位计时切断物料； 2. It is better to install a shut-off valve between the container and the level gauge to clean and repair the liquid level timer to cut off the material;
3. No conductive magnets are allowed near the main tube of the level gauge, otherwise it will directly affect the normal operation of the level gauge;
安装完毕后、需要用磁钢进行校正、对翻柱导引一次使零位以下显示红色，零位以上显示白色； 4. After the level gauge is installed, it is necessary to use magnetic steel for calibration, and guide the column once to display red below the zero position and white above the zero position;
5.When the liquid level gauge is put into operation, first open the lower drainage tube valve to allow the liquid medium to enter the main tube smoothly, to prevent the liquid medium from rising rapidly with the floating ball component, which will cause the column to turn black and the heart will fail or will turn randomly (if this occurs (After the rise and fall of the liquid surface is stable, it can be re-corrected with magnetic steel)
6. In order to avoid damage to the floating ball assembly during transportation, remove the floating ball assembly from the main tube of the small liquid level gauge before leaving the factory; after the liquid level gauge is installed, open the bottom drain flange, and then reinstall the floating ball assembly. Into the main tube, pay attention to the heavy end of the floating ball assembly facing up, and do not install it upside down;
7. According to the condition of the medium, the sewage discharge flange can be opened regularly to clean the sediment in the main pipe ...