First, the overview of the plastic tube rotor flowmeter
S-type plastic tube rotor flowmeter cone and other main components are made of AS, ABS plastic, which has good corrosion resistance. The product also has the characteristics of compact structure, small size, light weight, and the cone tube is not easy to be broken. Plastic tube rotor flowmeter can be widely used in products of industrial sectors such as chemical industry, environmental protection, food, etc ...
Second, the use of plastic tube rotor flowmeter
 When using the meter, first slowly open the upstream valve to full open, and then use the regulating valve downstream of the flow meter to adjust the flow; when the meter is not in use, slowly close the upstream valve; then close the flow regulating valve downstream of the meter;
 The indication value of the meter is seen in the middle of the score line, and the indication value is read from the scale of the float according to the height of the float;
 When the conical tube and float are contaminated, they should be cleaned in time;
 Regularly check the error of the flow meter. Once it exceeds the specified error, the float and cone tube should be replaced or recalibrated to obtain better accuracy;
 When the measured fluid is inconsistent with the calibration medium of the flowmeter, the displayed value should be corrected;
Installation instructions of plastic tube rotor flowmeter
 Before installation, the foam support in the instrument should be gently removed with tool nose pliers, and then reinstalled. When installing, pay attention to whether the rubber gasket is installed in the groove of the pipe to prevent it. Leaking
 The interface of the flowmeter can be matched with a plastic pipe, or with a plastic pipe, bonded with a special glue, or it can be connected by a plastic method, and then connected to a metal flange;
 The flowmeter must be installed vertically, without obvious tilt, and its installation height should be convenient for reading the float. When reading, the line of sight and the float should be horizontal; [General standard: The optimal flow rate required is the maximum flow rate of the flowmeter. 50% -70%, try not to use the highest number]
 There should be a straight pipe section 5 times the meter diameter at the upstream and downstream of the meter ...
 The pressure of the bedding fluid must be stable. If the fluid is too unstable, it will cause the float to fluctuate and cannot be measured correctly. A buffer or a setter should be set upstream of the flowmeter.
 The plastic tube rotor flowmeter should be installed vertically, and the angle between the centerline of the flowmeter and the plumb line should not exceed 5 degrees.
塑料管转子流量计的原理与结构 Principle and structure of plastic tube rotor flowmeter
The main measuring element of the plastic tube rotor flowmeter is a conical plastic tube with a small end facing downward and a large end facing upward, and a float that can move up and down inside. When the fluid flows through the conical plastic tube from bottom to top, a pressure difference is generated between the upper and lower parts of the float, and the float rises under the action of this differential pressure. When the force that causes the float to rise is equal to the combined forces of gravity, buoyancy, and viscous force, the float is in an equilibrium position. Therefore, there is a certain proportional relationship between the flow rate of the fluid flowing through the flowmeter and the rising height of the float, that is, the flow area of the flowmeter. The position height of the float can be used as a flow measurement.