| Address: Xiongyi, Jinhe Road, Jinhu County, Jiangsu Province |
| Phone: 0517-86900901 |
| Fax: 0517-86900902 |
| Postcode: 211600 |
| http://zzyfwh.com |
| Email: firstname.lastname@example.org |
| Features of Dongxiang input type liquid level transmitter |
| 2011-7-14 16:41:25 |
Features of the input level gauge produced by Jinhu Dongxiang Instrument Co., Ltd .:
The basic working principle of the input type liquid level gauge is the static pressure level measurement. In a liquid medium, the pressure at a certain depth is caused by the weight of the medium itself above the measurement point. It is proportional to the density of the medium and the local acceleration of gravity. The proportional relationship between them is reflected by the formula P = ρgh. Where P = pressure, ρ = medium density, g = gravitational acceleration, h = depth of the measurement point. Therefore, the physical quantity measured by the input type liquid level gauge is actually pressure, which can be understood by the calibration unit mH2O of the input type liquid level gauge . The actual liquid level must be obtained through conversion after knowing the two parameters of density and gravitational acceleration. Such conversion is usually carried out in the industrial field by a secondary meter or PLC.
The advantages of the input type liquid level gauge are: high measurement accuracy; easy installation; signals can be transmitted remotely; it can resist the corrosion of various media by choosing different materials; suitable for explosion-proof occasions; affordable price. The disadvantages are as follows: the measurement signal needs to be converted; the temperature of the high-temperature medium exceeding 125 ° C cannot be measured; the density of the measurement medium must be uniform.