The function of the glass rotor flow meter is to measure the flow
Low pressure loss, reliable performance, simple structure, convenient installation and use, cheap price
The main measuring element of the flow meter is a vertically mounted lower and upper conical glass tube and a float that can move up and down inside. When the fluid passes through the conical glass tube from the bottom to the top, a pressure difference occurs between the upper and lower parts of the float, and the float rises under the action of this differential pressure. When this rising force, the buoyancy and viscous lift force of the float are equal to the gravity of the float, the float is in an equilibrium position. Therefore, there is a certain proportional relationship between the flow rate of the fluid flowing through the flowmeter and the rising height of the float, that is, the area of the meter in circulation.
Introduction of glass float flowmeter
的主要测量元件为一根垂直安装的下小上大锥形玻璃管和在内可上下移动的浮子。 The main measuring element of the glass rotor flowmeter is a vertically mounted lower and upper conical glass tube and a float that can move up and down inside. When the fluid passes through the conical glass tube from the bottom to the top, a pressure difference occurs between the upper and lower parts of the float, and the float rises under the action of this differential pressure. When this rising force, the buoyancy and viscous lift force of the float are equal to the gravity of the float, the float is in an equilibrium position. Therefore, there is a certain proportional relationship between the flow rate of the fluid flowing through the glass rotor flowmeter and the rising height of the float, that is, the flow area of the glass rotor flowmeter. Glass rotor flowmeter is mainly used in chemical, petroleum, light industry, medicine, fertilizer, chemical fiber, food, dyes, environmental protection and scientific research and other departments to measure the single-phase non-pulsating (liquid or gas) fluid flow. Anti-corrosive glass rotor flowmeter is mainly used to detect the flow of corrosive liquids and gaseous media, such as strong acids (except hydrofluoric acid), strong bases, oxidants, strong oxidizing acids, organic solvents and other corrosive gases or liquids Flow detection of media.
Installation and use instructions
玻璃转子流量计是一种使用简单、读数方便、用途十分广泛的瞬时流量测量仪表。 How to choose glass rotor flowmeter correctly Glass rotor flowmeter is a kind of instantaneous flow measuring instrument which is simple to use, convenient to read and very versatile. In the field of environmental protection equipment and instruments, the amount is more than 30,000 units / year. Therefore, it is extremely important to choose and use this instrument. First, the type of glass to flowmeter and the selection of glass float flowmeter can be divided into: ordinary type, ribbed vascular type, micro flow and small outline type, corrosion resistance, laboratory type, Eight series of insulation type, alarm type and high pressure resistance type. According to the national series of instrument series spectrum, no matter which series includes a maximum of 12 calibers from 1mm to 100mm, the measurable flow range is: liquid (water) 0.1ml / min ～ 40m3 / hour, gas (Air) 1ml / min ~ 1000m3 / hour. The glass rotor flowmeter used for environmental protection instruments generally has a diameter of no more than 10 mm, and the measured flow belongs to a small flow range. The selection of glass rotor flowmeter can be considered from the following aspects. 1. The object of measurement. That is to measure the type of medium, pressure, and chemical properties. For liquid and gaseous media, for corrosive media, a corrosion-resistant flowmeter should be selected. 2. The performance of the flow meter itself. After the above conditions are determined, generally speaking, if there is no major change in price, the needle valve is preferred to be placed on the upper part of the flowmeter; those with larger flow holes are directly flow-scaled; simple structures; smaller external dimensions, etc . If it is a small flow range, you can use the ball float type, because it is stable during measurement, not easy to accumulate dust, high accuracy, and good interchangeability. 3． Select based on price. Generally speaking, high accuracy and high price. To select the accuracy level of the instrument according to the measurement purpose, if you only need to control the throughput of the measurement medium and adjust it after trial operation, you must always stabilize the throughput in the future, then the accuracy is secondary.
的刻度修正 玻璃转子流量计的刻度，是生产厂在本厂条件下用近于理想流体的水和干燥空气作介质标定得到的。 2. Calibration correction of glass rotor flowmeter The calibration of glass rotor flowmeter is calibrated by the manufacturer under the conditions of our factory using water and dry air which are close to ideal fluid. However, there are two situations where the scale value of the flowmeter cannot be used directly: one is that the measurement medium is not water and air, and the other is that the medium is water and air, but its state (temperature, pressure) and scale The status is different. In this way, when using a flow meter, in order to obtain a correct measurement result, a problem arises that the scale value needs to be corrected. Therefore, solving the calibration of the glass rotor flowmeter is the key to using this instrument. Considering that environmental protection instruments use rotor flowmeters to measure the flow of gaseous media, only the density correction in gaseous media measurement is discussed below. Because the viscosity of the gaseous medium is small, the viscosity effect is omitted in the discussion. Practice has proved that this does not affect the accuracy after correction