安装 Installation of electromagnetic flowmeter
1. The measuring tube must be kept full, and the installation position should preferably be at a low point. In the most ideal position, the fluid flows from low to high; 2. If it is placed horizontally, the two electrodes of the flowmeter must be kept on the same horizontal plane, in order to prevent sediment or air on the top of the tube to insulate it. 3. If necessary, the tube can be vented. The built-in empty tube detection circuit will stop the accumulation and flow signal output. For the fluid flowing in the anode-protected pipeline, the following points should be followed when installing the electromagnetic flowmeter: I. In order to ground the fluid well, we need to use a ground ring. II. When the sensor is connected to the pipeline, between the flanges If there is an insulation pad, an insulation sleeve must be used on the screw. III. The flanges on both sides of the pipe must be connected with 6mm2 copper core cables to prevent current from flowing through the sensor.
可以应用在不同的领域，如下：饮用水供水，补给水，水库和抽水泵站，等等。 Electromagnetic flow meters can be used in different fields, such as: drinking water supply, make-up water, reservoirs and pumping stations, and so on. Chemicals acids, lyes, conductive solvents, cooling liquids, additives, etc. Drink water, beer, wine, alcohol, milk, yogurt, juice, syrup, blood, etc. Cooling / heating (by an energy meter board) measures the energy used to cool or heat the air conditioner. Sewage sewage, industrial discharge sewage, domestic sewage, etc. Control of metal industrial pumps, cooling water, flowing water, etc. .Textile water, pigments, chemicals, bleach, etc.
的优点可测量任何导电率大于5S/cm的液体。 The advantages of electromagnetic flowmeter can measure any liquid with conductivity greater than 5S / cm. Not affected by the physical properties of the liquid being tested. Such as: temperature, pressure, viscosity. There are no mechanical moving parts, so no loss, no maintenance, no shrinkage, so no pressure loss. Used for sewage, water with particles is the most ideal. Measurable pipe diameter range (DN3 ... DN2000) very good linearity and high precision CIP certification, easy to clean. What you need to know about the selection of electromagnetic flowmeter General information? Type of liquid in the application area and its electrical conductivity Working conditions: working temperature and working pressure Lining material Electrode material Integrated or split measurement range Precision connection method Waterproof level